Q：Can we set some step as priority in the staining progress?
To handle the step in priority or control the exact stain time, we can set exact in the required step in the program setting.
Q：The loading station can not detect slide stands.
Please place the slide stands at the correct position and check the system.
Q：Is there an alarm system when any error detected?
DP260, DP261 andDP260S are provided with fault handing systems.
If an error is detected, the detailed error message and corresponding instructions of user's operation are displayed on the touch screen.
Q：Can we use the slide stainer for frozen section?
The stainer has good compatibility, can be used for H&E staining and Papanicolaou staining, and Frozen section staining.
Q：How can we change the exit station into entry station?
Click the target station in main interface,
Click change station in the popup,
Select the entry station in the pull-down list,
Save the setting.
Q：How long time the activated carbon filter should be changed?
Depending on the use frequency, commonly 3-6 months we should change the carbon filter.
Q：To avoid the reagents evaporation, can we shut down the waste gas exhaust system?
The fan can be set as open, intelligent and shut down modes for different needs.
In open mode, the fan keeps running.
In intelligent mode, the fan will be started according to the setting value.
In shut down mode, the fan will be not running.